for your patient and your profession
How to clean
Rinse the device under warm running water for a minute or use an ultrasonic cleaner, away from edgy instruments and sharp. Wipe with gauze or a soft cloth, taking care not to scratch the surface.
How to sterilize
The product can be sterilized in an autoclave at 135 ° C for 10 minutes or with a UV lamp for 60 minutes.
Check the surface and mirror's edge looking for any non-uniformity. Do not use the device if the edge or the surface are damaged. Moreover do not use the mirror if you notice discontinuity or chipping on it that could hurt the patient and / or operator.
Before using it, to prevent fogging, pass the mirror under warm water for a few seconds.
Dry with a soft cloth or gauze.
"Q: Which are the certified sterilization methods?
R: Sterilization in the autoclave at 135 ° for 10 minutes or with a UV lamp for 60 minutes "
"Q: Could you create custom shapes to be used only for our company?
A: We manufacture any custom shape "
"Q: Your product is in compliance with the law?
A: Our product is certified ISO 17664 and CE, and complies with FDA 510 (k) "